Time of last reading:
Aug. 16, 2018, 6 p.m.
Jan. 25, 2018, 8 p.m.
Issue Feb. 12, 2018 - Feb. 22, 2018
It has been brought to our attention by Environmental Instruments Ltd (manufacturers of AQMesh) that due to a bug in a server systems update that NO2 and O3 data from all AQMesh pods were effected from 7am on 12th February until 9am on 22nd February 2018. No other channels are affected. We caution you against using NO2 and O3 data for this period until the corrections have been applied. We urge users to re-call this data from Tuesday 27th when all readings will have been re-processed. The Urban Observatory apologises for the inconvenience.Issue April 6, 2018 - April 6, 2018
NO data reading negative. Sensor needs replacing. NO2 data reading too high. Sensor needs replacing.Issue April 9, 2018 - April 9, 2018
Replaced faulty N02 sensor at Corner House AQ MeshIssue May 1, 2018 - May 1, 2018
Aq_meshes 1904 (Chillingham Rd Primary School) , 1905 (Corner House, Heaton) removed for slope/offset adjustment at Precision SiteIssue May 22, 2018 - May 22, 2018
Deployed Coast Road Corner House aq_mesh1971150 as replacement for aq_mesh1905150 Lamp post 8S235 Post Code NE6 5QD GPS 54.99113938 -1.583788097Issue May 22, 2018 - May 22, 2018
Chillingham Road Primary (1904150) and Coast Road Corner House (1095150) were not 'refurbished' back in Jan 18 with the rest of that batch. These have been removed and substituted for new AQ Mesh pods ie Chillingham Road Primary is now aq_mesh1970150 Coast Road Corner House is now aq_mesh1971150 . Please refer to those pods from this date till further notice. ThanksIssue Aug. 2, 2018 - Aug. 2, 2018
Redeployed ( formerly SenseMyStreets Coast Rd/Corner House) Now at St Teresa's Primary School Heaton Rd Lamppost 27H116 NE6 5HN 54.984003 -1.58199
Sensor Maintenance Records: