Time of last reading:
July 16, 2018, 9 a.m.
Dec. 20, 2017, 4:45 a.m.
Issue Feb. 12, 2018 - Feb. 22, 2018
It has been brought to our attention by Environmental Instruments Ltd (manufacturers of AQMesh) that due to a bug in a server systems update that NO2 and O3 data from all AQMesh pods were effected from 7am on 12th February until 9am on 22nd February 2018. No other channels are affected. We caution you against using NO2 and O3 data for this period until the corrections have been applied. We urge users to re-call this data from Tuesday 27th when all readings will have been re-processed. The Urban Observatory apologises for the inconvenience.Issue March 19, 2018 - May 23, 2018
Gosforth Gold aq_mesh1917150 did not report since 19th March.Issue May 23, 2018 - May 23, 2018
Battery replaced and solar panel re-oriented to southwardsIssue June 10, 2018 - June 27, 2018
NO2 sensor reading high 1000s from 10/06/18 Replaced on 27/06/18 Failed NO2 sensor: 202280703 New NO2 sensor:202581451
Sensor Maintenance Records: